Why send your UTI ashore, when you can service it yourself?
The Bruusgaard UTI solution is based on the SHT 2002. It is designed to measure ullage, air/oil/water interface levels, and their respective temperatures – in one single operation.
What really separates this UTI solution from the rest, is its onboard calibration capability. There’s no need to send your equipment ashore for servicing. We’ve included everything you need to operate, service and calibrate your SHT2002 in the service kit.
Preparing the SHT UTI 2002 is straightforward, but nevertheless important.
The unit is ready for use when it displays the temperature.
The SHT2002 is now ready for use.
The UTI maintenance and service kit contains everything you need to perform service on board:
In the section, you’ll find a step by step guide on how to service and calibrate the unit.
Periodic maintenance and service is necessary for delicate equipment such as the SHT2002. This is how you perform the basic overhaul.
When you have replaced the parts, one final task remains. The SHT2002 requires an annual calibration to ensure that the device functions properly.
The first thing you check is the temperature.
If the temperature readings deviate more than 0.2 degree Celsius, replace the sensor.
You then proceed to check that the sensor responds to interface changes.
If the beeping pattern does not reflect the interface, replace the sensor.
Having completed the calibration, the SHT2002 is all set. Good luck!